Most Active Stories
- Non-Verbal & Non-Linear, Jill Burton’s Singing Creates Spontaneous Soundscapes
- Choral Arts of Chattanooga Presents 'Wonderful Peace' Holiday Concert 12/13
- 'A Christmas Story, The Musical' Adds New Depth to Beloved Characters
- Give a meaningful gift through WUTC
- UTC recognized by U.S. News and World Report for work with student veterans
Shots - Health News
Fire Risk Leads Praxair To Recall Grab 'n Go Oxygen Tanks
Originally published on Thu January 3, 2013 2:05 pm
When there's a cylinder of medical oxygen in the room, the last thing you want it to do is burst into flames.
So the Food and Drug Administration says Praxair has recalled its Grab 'n Go Vantage portable oxygen units. The product combines an aluminum cylinder with a regulator that releases the oxygen at the right pressure for medical use.
It's a handy combination, but only when the Grab 'n Go is working right.
Turns out that if these cylinders get kicked around or knocked over they can sometimes catch fire.
The flames can cause the cylinders to break open, according to the FDA. People nearby could get burned. Patients would also be at risk of not getting enough oxygen.
Previously, Praxair said its investigation found that Grab 'n Go units with a particular type of o-ring between the regulator valve and the cylinder "may be more prone to this type of incident."
When the cylinders are returned for servicing or refilling, Praxair is replacing the old o-rings with Teflon models. Praxair has told customers to send the tanks back for the fix and to be careful in handling them.
As of late last month, about half of the units in the U.S. had been retrofitted. The ones that have been fixed carry a decal reading "T" for Teflon.
This is a Class I recall, which is the most serious kind in the FDA's system. That status is reserved for cases where use of the product could cause serious health problems, including death.